Nonin PureLight® Technology
For accuracy at critical SpO2 levels.
SpO2 Low Perfusion Accuracy: 70 to 100% ±2 digits (Arms*)1
Pulse Rate Accuracy: 40 to 240 BPM ±3 digits (Arms*)1
Pulse Rate Low Perfusion Accuracy: 40 to 240 BPM ±3 digits (Arms*)1
Temperature: 4, 5
Operating: -20 °C to 50 °C (-4 °F to 122 °F)
Storage/Transportation: -40 °C to 70 °C (-40 °F to 158 °F)
Humidity: 4, 5
Operating: 10% to 95% non-condensing
Storage/Transportation: 10% to 95% non-condensing
* ±1 Arms encompasses 68% of the population at zero bias.
1 Additional accuracy and performance information can be found in the sensor accuracy document on the operator’s manual CD.
2 Accuracy testing was performed under no-motion conditions.
3 Accuracy specifications based on Nonin’s PureSAT® SpO2 technology and PureLight® sensor technology.
4 For combined oximeter/sensor specifications, refer to the applicable oximetry system’s operator’s manual.
5 Range as tested with Nonin’s PureSAT SpO2 technology.
Measurement Wavelengths and Output Power**:
Red: 660 nanometers @ 3 mW nominal
Infrared: 910 nanometers @ 3 mW nominal
** This information is especially useful for clinicians performing photodynamic therapy.
This product complies with ISO 10993-1.
Not made from natural rubber latex.
Can be contoured to fit on any patient
For durability and easy cleaning of sensors
Hybrid sensor system saves money over traditional disposable sensors
Trim design for a great fit; not made with natural rubber latex